This Circuit helps in the monitoring of mains supply voltage. It does not use a isolation step down transformer. This has to be constructed only by skilled people with knowledge of safety requirements.
C1 0.47uF can be brought down to 0.22uF for low LED currents, use high efficiency ultra bright LEDs.C1 should be 440V AC or 630V DC plastic axial yellow, polyester, polycarbonate, polypropylene, metalized film.
R3-R6-R9-R14-R18 resistor divider determines the LED turn on or threshold switch points, 10M for hysteresis.
Mains Voltage and Power Circuits
Adjust R16 preset with a log Plastic tweaker to get the led D2 to turn on when input voltage is at 220V AC. This has to be done after PCB is put in a sealed fire-retardant-plastic or epoxy box. drill a hole in box for plastic tweaker.
LM324 – Low Power Quad Operational Amplifier
This Circuit is Not a Tested Design. It is an Idea for study
Warning : This Circuit is Mains Operated without Isolation Transformer and will give lethal electric shock if touched when the circuit is turned on. Test circuit only with DC 9V Bench Power Supply to try it out. Do not use 230V AC.
LM331 is a Voltage-to-Frequency Converters and 74C926 is a 4 Digit Counter. The first part of the circuit converts the amplified DC analog value from a shunt to pulses. The Second counts the pulses and Displays it on a Seven segment LED display.
In Electroplating, the time taken will determine the plating thickness which is both related to cost and quality. More current also means more metal has been used so it has to be monitored to control and deliver cost effective electro plating quality.
Mixed and Interface Circuits
A high current with low voltage is applied on special electrodes. One of the electrodes will have the job to be plated. The Medium is an Electrolyte. This is a part of Electrochemistry, Battery, Cells, Electrolytic Capacitors all belong to this branch.
When Instruments are designed a analog front end is essential and also as most equipment have digital or microcontroller interface the analog circuit needs to have digital access. The Circuits DACT0008 and DACT0009 are both useful in building instruments which have digital control.
Precision Attenuator with Digital Control
The Circuit DACT0008 is a programmable attenuator and the digital control can be a remote dip switch, a CMOS Logic Output like the A-B-C-D outputs of a decade counter, or an I/O port of a uC like 80C31.
The heart of the circuit is the popular OP07 OpAmp with Ultra Low Offset in the inverting configuration, 4052 a CMOS analog multiplexer switch enables the gain change, the innovation of the circuit is that the on resistance ( around 100 ohms) of 4052 switch is bypassed so that no error is introduced by its use.
The resistors used R1 to R6 can be 0.1% 50ppm if you will use a 3 ½ DPM i.e. + /- 1999 counts ( approx. 11 bit ), but for 4 ½ DPM ( approx. 14 bit ) you may need to have trimpots2 in place of R3, R4, R5 & R6 gain selection resistors to properly calibrate to required accuracy but for testing or trials use 1% 100ppm MFR resistors but the errors will be around 1%.
Output connect to DPM 7107/7135 or any other A/D Convertor or OpAmp Stage. Use a buffer at output if output has to be loaded by a value less than 1Meg. Use an inverting buffer if input leads have to have polarity where gnd is -In. See DACT0009 for details.
c. 4052 CMOS Switch
The 4052/51/53 Analog Multiplexers have an on Resistance of around 100E the highlight of the circuit is that the CMOS on resistance comes in series with the opamp output source resistance, which produces no error at output.
Digital Control Options
A and B can be controlled by I/O port of uC, like 80C31 so that the uC can Control gain. A and B can be given to Counters like 4029/4518 to scroll gain digitally. A and B can be connected to DIP switch or thumbwheel switch.