These are the circuits and boards of a Mini Temperature Transmitter for a Platinum hundred ohms temperature Sensor. Has a 4-20mA sink output. Current Loop can be used for multiple instruments.
Mini RTD Pt-100 Three Wire Transmitter
The customer wanted a two wire system, this needs low power opamps which were ten times the price. If this transmitter works at 3mA it could have been two wire. Those parts were in short supply and the budget of customer was low. So i made it with regular opamps but three wire. The customer was satisfied as he got a cost effective solution. But now the situation of markets and products are different.
If you need to transmit the temperature from a high voltage area or toxic environment. You will need to hermetically seal the transmitters, convert the Voltage to frequency, frequency to IR LED flashes or RF. This can be then remotely monitored. Then the transmitter has to be battery-solar operated, low power too. ICL7135 is a simple solution. It has a serial output that can be used to Drive IR Leds. Remotely sense these flashes in a Micro-controller and you have a reading. You may be able to use optic-fibers too. Where volatile liquids are present so that the risks of sparks can be eliminated.
RTD Mains Power Transmitter
This is the same Mini RTD Pt-100 Transmitter but in its case. Encapsulated in epoxy, hermetically sealed against harsh industrial environment. This will work well even near fumes of Ammonia with no corrosion. But not near vapors which can be ignited by sparking as terminals are still open. The side view shows zero and full scale ten turn bourns trimpots heads, for calibration. After cal it can be sealed with RTV compound.
U1 CD4541 is a Timer with Long Duration Ability. This timer is started by a low pulse from earlier circuit, when the temperature goes above setpoint, a low state is at U2A inputs, this starts the timer. The timer output goes high after a preset time. U2C-D Flip flop is power on reset via cap C7. Even a manual ‘reset’ is used if required. This alarm toggle is ‘set’ when The temperature remains high even after the preset time period of U1.
Timer and Counter Modules – del90002
A digital counter monitors the number of times a process goes beyond a certain temperature limit over a long cycle of time. This circuit has a Analog Timer and Digital Counter. The Analog timer turns on an Alarm if the Temperature Limit transition is very long.
Here 4052 is used as an analog multiplexer, U3A TL062 opamp is wired as a 1mA Constant Current Source. It pumps 1mA into U2 4052 pin 13 X . Depending on BCD code on inputs A, B of 4052 the current is routed to any one of the four RTD 100E, whose one end is connected to X0-X3. The current then flows to GND thru the Platinum 100E resistance. The Y0-Y3 monitors the mV developed on RTD in tandem with X0-X3 positions. Like a ganged rotary switch. The Output mV at Y is the mV of Active Channel as selected by the BCD of 4029.
RTD Pt-100 Four Point Alarm – del90001
The BCD is provided by 4029 counter which is clocked by a 555. U5A cancels out the 100E mV (1mA * 100E) of a cold RTD and Amplifies the differential mV. This output of U5A is in proportion with measured temperature. U5B compares the mV that was obtained with a preset mV of POT1, which is a user setting called setpoint. The difference is amplified by U5B which is saturated by U3B comparator which adds a little hysteresis too. R7-C2 further dampen and slow the response. This finally drives Q2 to provide a Logic Signal indicating if temperature is below or above setpoint.
Four points in a Closed Loop Temperature Control System is Monitored and a Alarm set to go off when the temperature goes beyond the set limit.