# Ammeter and Precision Rectifier

Studying current measurement is a prerequisite for many of the measuring techniques. The current parameter mainly specifies the power consumption in a circuit, given the value of resistance. It is found convenient to measure current rather than voltage for knowing power output and determining efficiency. It may be required to measure leakages in circuits at certain times. Hence the measurement of current constitutes a priority.

Measurement of DC Current –

The circuit diagram for the measurement of current (d.c. and a.c. modes) is shown aside. For measurement of current switch SI is operated. The switch S-ad is kept in d.c. mode. This enables the current to pass through a shunt circuit consisting of resistors R26, R27, R28, R29 and R 30. The current ranges are provided in 5 decades i.e. 200 micro-amps, 2 milli-amps, 20 milli-amps, 200 milli-amps and 2 amps. An additional current range that can be read upto 20 Amps is also provided. However, for measuring this high current the green terminal provided on the meter should be used. When a current to be measured is fed to the input terminals of the instrument appropriately, a voltage proportional to the current through the shunt resistor is fed into the DPM which measures the d.c. voltage which in turn indicates the d.c. current being fed.

Measurement of AC Current –

In case of a.c. measurement, the switch S-ad is kept in a.c. mode. The a.c. current path is similar to the d.c. current path in the shunt resistor. However the voltage tapped across the shunt resistor is fed into IC2 which is a buffer. The output of IC2 is fed to IC3 through capacitors C10 and C11. This IC is an operational amplifier acting as a precision rectifier. The output of IC3 is fed to the input of the DPM for measuring the a.c. current being fed to the input terminals. It can be seen that the current measurement is similar to the voltage measurement except that the attenuator chain is replaced by the shunt resistor circuit.

(This is scanned-ocr from my earlier file, some mistakes corrected – delabs)

• Dual Polarity Power Supply
This supply gives both positive and negative outputs. Appropriate Fuses should be used to protect from fire hazard and overload of transformer. Voltage Regulators LM7812 [...]
• 0-1V to 4-20 mA Converter
This two opamp circuit converts analog voltage signals to current (sink) signals in a proportional manner. Current signals are more immune to noise and cross [...]
• Edison Bulb Life Extender
The Edsion Filament lamp may be less popular, but it is the most cost-effective light even today. It is very affordable, The light is soothing [...]
• Current Source for Resistance Measurement
Here is a current source you can build for resistance measurement. When the current is held constant, you know as per Ohm’s Law the Voltage [...]
• Interface ICL7135 Voltmeter to Atmel 89C52
This circuit is a Parallel interface between 89C52 of Atmel with 7135 of Intersil.  With This circuit you can read analog data of both polarities. [...]
• Mains Voltage monitor using LM3914
This Circuit helps in the monitoring of mains supply voltage. It does not use a isolation step down transformer. This has to be constructed only [...]
• Battery Level LED Indicator
This LED Indicator uses a LM339, a quad comparator. LM339 can work on single or dual supplies, it has a open collector output that can [...]
• WorkBench Dual Power Supply
I will just explain part of this circuit. D9 and D10 provide a low cost -1.4 from -5 V. This is needed to reach near [...]
About delabs
Product Design - Industrial Automation and Instrumentation. -