This is a LED Analog Meter, This can be used as a Resistance Meter and Low Impedance Voltmeter for Battery Levels. To measure battery voltage, the R5-R12-R17 etc. part of the Reference Resistor Divider Network can be modified to suit. Shown here is for 4 LEDs, Use Three LM324 for 12 or More LEDs and Cascade as shown.
This cannot Measure Voltage levels from High Impedance Sources, will work for Battery Voltage Tests. To make it into a Continuity tester. R27 must be a short and R23 5 Ohms. The Black probe should have a Built in Resistance of 2 Ohms. If you want it to be a dedicated voltmeter, remove R3, The Probe has to be a 10X Attenuator with 10M Ohm and The Resistor Divider Steps in 100mV per Step. The R27, R23 etc. is 20K. A Leakage Tester a Mains Voltage Monitor are other possibilities. Use LM3914 for a easier solution. A nice book for your Design Library – Measuring Circuits By Rudolf F. Graf
This is easy to rig millivolt source for field calibration or troubleshooting of 4-20 mA current loops. Here a Darlington pair is used for current amplification which reduces the Ib error as gain is very high.
A rotary switch selects, 4-12-20 mA Preset points. A Bourns multi-turn wirewound Pot can also be used with a digital dial. Enclose in a dust proof handheld box. Read more on process calibration.
When a Inverting Opamp Configuration is at a steady state, we say the Inverting Input is at a Virtual Ground. That means it is at 0V w.r.t to the dual power supply ground, but it cannot drive or draw any current. It is at a high impedance, but still at 0V. When you buffer this 0 V, you get a low signal ground for a opamp supply.
This gnd. can sink and source in a couple of mA. You can use it with low power opamp circuits for portable battery operated devices. This creates a virtual +/- 6 V dual supply from a 12V battery. This may be needed in cases where some instrumentation opamps need the negative supply or your design demands a measurement around zero. You may get a more loadable ground using a Power Opamp, i have not tried. The above circuit gnd cannot be used as a return path for LED’s or Relays. You can drive these, between VCC-VDD, but translate levels to drive them.
This was a small circuit made for driving an Impact counter. The heart being ICL8038. It must have been a Motor driving a Conveyor, the motor has a feedback attachment called Tachogenerator.
Only part of the circuit is shown here. See the image of product here Tacho Counter. The configuration is derived from the Application Notes of Intersil. The Voltage from Tachogenerator is Measured on a DPM-DVM and also fed to this circuit after attenuation and filtering. The square pulses of 8038 is used to derive a Logic pulse train for a CD4040. The CD4040 works of 0 and 12V. The above circuit is on +12 and -12. That is the reason R10-R15-Q1 are used. The pulses are 0-12V pulses. The Zero is Virtual like the Virtual Ground in low current power supplies.