Here 4052 is used as an analog multiplexer, U3A TL062 opamp is wired as a 1mA Constant Current Source. It pumps 1mA into U2 4052 pin 13 X . Depending on BCD code on inputs A, B of 4052 the current is routed to any one of the four RTD 100E, whose one end is connected to X0-X3. The current then flows to GND thru the Platinum 100E resistance. The Y0-Y3 monitors the mV developed on RTD in tandem with X0-X3 positions. Like a ganged rotary switch. The Output mV at Y is the mV of Active Channel as selected by the BCD of 4029.
RTD Pt-100 Four Point Alarm – del90001
The BCD is provided by 4029 counter which is clocked by a 555. U5A cancels out the 100E mV (1mA * 100E) of a cold RTD and Amplifies the differential mV. This output of U5A is in proportion with measured temperature. U5B compares the mV that was obtained with a preset mV of POT1, which is a user setting called setpoint. The difference is amplified by U5B which is saturated by U3B comparator which adds a little hysteresis too. R7-C2 further dampen and slow the response. This finally drives Q2 to provide a Logic Signal indicating if temperature is below or above setpoint.
Four points in a Closed Loop Temperature Control System is Monitored and a Alarm set to go off when the temperature goes beyond the set limit.
When Instruments are designed a analog front end is essential and also as most equipment have digital or microcontroller interface the analog circuit needs to have digital access. The Circuits DACT0008 and DACT0009 are both useful in building instruments which have digital control.
Precision Attenuator with Digital Control
The Circuit DACT0008 is a programmable attenuator and the digital control can be a remote dip switch, a CMOS Logic Output like the A-B-C-D outputs of a decade counter, or an I/O port of a uC like 80C31.
The heart of the circuit is the popular OP07 OpAmp with Ultra Low Offset in the inverting configuration, 4052 a CMOS analog multiplexer switch enables the gain change, the innovation of the circuit is that the on resistance ( around 100 ohms) of 4052 switch is bypassed so that no error is introduced by its use.
The resistors used R1 to R6 can be 0.1% 50ppm if you will use a 3 ½ DPM i.e. + /- 1999 counts ( approx. 11 bit ), but for 4 ½ DPM ( approx. 14 bit ) you may need to have trimpots2 in place of R3, R4, R5 & R6 gain selection resistors to properly calibrate to required accuracy but for testing or trials use 1% 100ppm MFR resistors but the errors will be around 1%.
Output connect to DPM 7107/7135 or any other A/D Convertor or OpAmp Stage. Use a buffer at output if output has to be loaded by a value less than 1Meg. Use an inverting buffer if input leads have to have polarity where gnd is -In. See DACT0009 for details.
c. 4052 CMOS Switch
The 4052/51/53 Analog Multiplexers have an on Resistance of around 100E the highlight of the circuit is that the CMOS on resistance comes in series with the opamp output source resistance, which produces no error at output.
Digital Control Options
A and B can be controlled by I/O port of uC, like 80C31 so that the uC can Control gain. A and B can be given to Counters like 4029/4518 to scroll gain digitally. A and B can be connected to DIP switch or thumbwheel switch.