This was designed, keeping in mind, that sometimes we have few components available and you could not carry your DMM somewhere, but you have to make some instrument for a quick need with what is available.
Analog Dial AVO Amps-Volts-Ohms meter
IC3 LM555 is to generate -5V from 9V. That reminds me of a story, In 1986 while i was studying BE electronics i got a chance to work part-time in a company that sold-serviced imported instruments, it was called trans-marketing. They were the agents for Racal-Dana, Genrad, Data I/O and so many more. Here i came across a Book called CMOS Hot Ideas of Intersil. I had a chance to observe, build and learn some things here. The ICL7107 circuit was the most fascinating, I also read the books of National Semiconductor here.
In those days small firms used to make DPM’s digital panel meters with 7107. In 1987 when i got a oppurtunity to design a DPM for a firm, I put the 555 clock in place CD4009 clock shown in intersil, to derive the -5.
LM331 is a Voltage-to-Frequency Converters and 74C926 is a 4 Digit Counter. The first part of the circuit converts the amplified DC analog value from a shunt to pulses. The Second counts the pulses and Displays it on a Seven segment LED display.
This Circuit is a DPM or digital panel meter. It has a analog bar graph display and a 3-1/2 digit digital display. ICL7107 is used in the 200mV configuration.
U4A opamp LF353 amplifies the 200mV Full scale input to the level required for the LM3914 display circuit. D13-D14 are clamping protection diodes. Adjust P1 trimpot for a reading of 1000 counts when a 100.0 mV signal is fed at Vin. Adjust R8 trimpot to get the 5th led to just turn on at 100mV input.
Analog and Digital Voltmeter using ICL7107
A combination of digital and analog display is helpful for quick decision making. Analog indicates even from a distance the process dimension.
Human Brain understands analog better. The digital is required to note down and record values for determining a setpoint or performance of a system.
Sometimes a analog recorder with a ink-pen plot against time is a very good way of process analysis. Many systems are better studied using graphs not tables and lists of numbers. A Computer based data-logger gives greater power to this methods. You can measure and plot graphs of various types and at different points and for much longer periods.
This is a part of my Build a DMM or Digital Multi Meter
Here is a easy to read ‘Analog’ Millivoltmeter. Just like the Moving Coil Voltmeter, but does not have that resolution. This gives a easy indication of process progress or parameter magnitude from a large distance. A bargraph is easy on decision making too, compared to a digital readout.
How this Works ? – The analog input in mV – millivolts is fed to R18, RC reduces Noise and the Zener Clamps protect. The LF353 FET Opamp offers High Impedance as a Non-Inverting Amplifier, which nullifies measurement burden, Remember the Moving coil voltmeter loads the measured circuit, causing sizable errors. Then they invented the Vacuum Tube Voltmeter to solve this issue. An FET voltmeter is near ideal, they ought to have inventing this first.
The Zeners and C3 Plastic cap should not leak, even if they do it should be in Pico Amps. Get quality stuff and do a neat job putting them together. To master this leakage and other aspects, try building an Electrometer with CA3140. Another Measurement challenge is uV Microvolt measurements, you will be faced with new glitches in connectors and PCB due to thermocouple effects and contact resistance. A soldered joint near a hot resistor will set up enough thermal gradients and create many thermocouples all over the board. Try to measure 1 Microohm with a 10 Amp pulse or 1 A DC. You will learn many things. Connectors have a craze for the Precious Metal, they act funny if they do not have enough Gold on em.
Once i observed, very low voltages or circuits with nominal voltages but very low currents, cannot break a near invisible layer between the plates of a good connector. A sub-micron coat of corrosion, dust or even some organic deposit, was forming a dielectric layer which was impervious to uV and pA. A good cleaning with a volatile organic solvent solved the problem but messed up other plastics nearby.
Millivolt Meter using a LM3914 LED Dot Display. – This circuit is a part of my Build a DMM or Digital Multi Meter