I don’t remember if this circuit worked properly. But a few were made and i might not have shown the modifications that were done to make it work. This was meant to be a portable, low cost, insulation tester for an electrician. If you try it out and debug it it may work well.
High Resistance Insulation Materials
A negative voltage is derived by shifting gnd with two diodes, i feel this did not work very well. Two pins of CD4028 pins are also used to boost the reference to get two extra ranges as 4051 has a 100E on resistance.
The 555 clock makes 4029 counter count. But the clock can be clamped to gnd by a TL062 window comparator. The clock is frozen when the input value to comparator pin 5-2 is within a lower limit and upper limit “window” pin 3-6.
The 4029 counter BCD is decoded to decimal by 4028 which drives the LEDs, keep LED drive within 3mA or chip will be loaded. Use high efficiency extra-bright LEDs.
The 4029 BCD also controls a shunt resistor array with CMOS switches 4051. The voltage across shunt is a sample of leakage current. This is compared in the window comparator to freeze the Clock and LED display to give a reading of the leakage current or Insulation Resistance.
LM331 is a Voltage-to-Frequency Converters and 74C926 is a 4 Digit Counter. The first part of the circuit converts the amplified DC analog value from a shunt to pulses. The Second counts the pulses and Displays it on a Seven segment LED display.
In Electroplating, the time taken will determine the plating thickness which is both related to cost and quality. More current also means more metal has been used so it has to be monitored to control and deliver cost effective electro plating quality.
Mixed and Interface Circuits
A high current with low voltage is applied on special electrodes. One of the electrodes will have the job to be plated. The Medium is an Electrolyte. This is a part of Electrochemistry, Battery, Cells, Electrolytic Capacitors all belong to this branch.
This is a Test Circuit for Logic Analyzer software, with the Simple port circuit. The VB Logic Analyzer works, but is just an example program to test and learn.
The speed at which the logic state changes say like 10 M Hz while probing even a PIC or 89C51, is too fast for this circuit. For that, the incoming data has to be spooled or stored in RAM at real time, by the external Hardware itself.
Test Circuit for Logic Analyzer software – del20006
As this is a Parallel Port Interface, an Embedded High Speed Digital RAM Storage of Multiple Logic input Channels with Signal Conditioning in the front end, could do the job. This is the minimum, if you want to program the rest of the instrument on Computer Software. Here is one you can see from Bitscope using USB or Ethernet.