U1 CD4541 is a Timer with Long Duration Ability. This timer is started by a low pulse from earlier circuit, when the temperature goes above setpoint, a low state is at U2A inputs, this starts the timer. The timer output goes high after a preset time. U2C-D Flip flop is power on reset via cap C7. Even a manual ‘reset’ is used if required. This alarm toggle is ‘set’ when The temperature remains high even after the preset time period of U1.
Timer and Counter Modules – del90002
A digital counter monitors the number of times a process goes beyond a certain temperature limit over a long cycle of time. This circuit has a Analog Timer and Digital Counter. The Analog timer turns on an Alarm if the Temperature Limit transition is very long.
Here 4052 is used as an analog multiplexer, U3A TL062 opamp is wired as a 1mA Constant Current Source. It pumps 1mA into U2 4052 pin 13 X . Depending on BCD code on inputs A, B of 4052 the current is routed to any one of the four RTD 100E, whose one end is connected to X0-X3. The current then flows to GND thru the Platinum 100E resistance. The Y0-Y3 monitors the mV developed on RTD in tandem with X0-X3 positions. Like a ganged rotary switch. The Output mV at Y is the mV of Active Channel as selected by the BCD of 4029.
RTD Pt-100 Four Point Alarm – del90001
The BCD is provided by 4029 counter which is clocked by a 555. U5A cancels out the 100E mV (1mA * 100E) of a cold RTD and Amplifies the differential mV. This output of U5A is in proportion with measured temperature. U5B compares the mV that was obtained with a preset mV of POT1, which is a user setting called setpoint. The difference is amplified by U5B which is saturated by U3B comparator which adds a little hysteresis too. R7-C2 further dampen and slow the response. This finally drives Q2 to provide a Logic Signal indicating if temperature is below or above setpoint.
Four points in a Closed Loop Temperature Control System is Monitored and a Alarm set to go off when the temperature goes beyond the set limit.